Practical operation skills of ultrasonic cell crusher homogenizers
Ultrasonic cell crusher is also named ultrasonic homogenizer, ultrasonic sonicator, ultrasonic disperser, ultrasonic
emulsifier, it is widely used for Emulsification, Separate, Dispersion, Homogenize, Extract, Dega, Cleaning, Speed
up chemical reaction. Below is some practical operation skills of ultrasonic cell crusher homogenizers.
1. Please be careful not to turn on the ultrasonic work when the ultrasonic probe is not inserted into the sample,
otherwise the noise will increase in the subsequent use. If the ultrasonic start to work without inserting the sample,
the longer the time, the greater damage of ultrasonic emulsification.
2. The water inlet depth of the ultrasonic probe of the ultrasonic emulsifier is about 1.4cm, and the height of the
liquid level is preferably above 30mm. The ultrasonic probe should be centered without being attached to the wall.
Because the ultrasonic wave is a vertical longitudinal wave, it is not easy to generate heat convection if it is inserted
output power and selection of reaction vessels. The ultrasonic time should not be too long each time, preferably no
more than 5 seconds, and the gap time should be greater than or equal to the ultrasonic time, which is good for the
release of heat. The output power is not suitable for large, otherwise the sample will splash or foam. Of course,
suitable reaction vessels are also beneficial to the thermal convection of the sample in ultrasound, and improve the
high efficiency of crushing.
4. In daily maintenance, clean the ultrasonic probe with ethanol or turn on the ultrasonic with cold water after use.
Various type ultrasonic homogenizer customization, OEM & ODM, factory directly sale to help you save big, in stock
fast delivery, contact email@example.com to enquiry now.