it's a multi-function and multi-purpose instrument to do ultrasonic treatment.
Ultrasonic emulsification dispersion machine is also popular as ultrasonic homogenizer and lab homogenizer, which
is applied in plant & animal tissues, cells, strains of bacteria, spores broken, at the same time could be used to emulsify,
separation, dispersing, homogenization, extraction, degassing, cleaning and accelerate chemical reaction, etc.
1. Setting temperature protection temperature number must be higher 1-3℃ than room temperature or sample temp.
2. Strictly prohibit power on the instrument when amplitude transformer not insert in liquid (no-load situation), or will
3. For different sample quantity, different time long or short, different power, need customer ensure these according
different samples, and then choice best parameters.
4. After using a period time, the end of amplitude transformer maybe cavitation and beard hair, you could file flat by
oil or file, or will effect working effection.
5. When ultrasonic crushing ( ultrasonication ), the liquid temperature will raise as cavitation effect, so you should
pay attention to different cellular temperature.
6. Short time (not over 5s) many times crushing is recommend, and at the same time could add ice bath to cooling
at the outside. (Suggest ultrasonic working 1-4s, interval 2-8s working).
7. Warning: During working, if you want to change amplitude transformer, must power off, and re-choice amplitude
transformer type when pow on again, or will damage amplitude transformer.
With ISO & CE certification high quality guarantee, through TOPTION engineer professional design and quality
performance, large screen LCD display and central microcomputer centralized control,etc. Ultrasonic emulsification