How to use rotovap distillation to better cbd extracts?
In order to produce highest-qulity cbd oil, researchers are looking to more traditional methods of hemp
extraction and isolation, in fact, there're several options to improve plant oil manufacturing techniques.
Rotary evaporator is mainly used as an economical solvent removing / ethanol recovering equipment during
hemp oil distillation process. At here let's learn together, how to use rotovap distillation to better isolate the
cbd extracts and components.
Four main ways to make your rotovap disitllation process faster:
- Increasing the evaporation flask size.
- Increasing the evaporation flask rotating speed.
- Increasing the rotary evaporator's bath temperature.
- Increasing the vacuum level.
1.Increasing the evaporation flask size.
Same as we know, the rotary evaporation flask should never be more than half full, both to prevent spillage
and allow enough liquid surface area for efficient evaporation. So when we choose rotary evaporator capacity,
this point must be considered.
For laboratory research, choose a small rotary evaporator is more economical and practical. But I also want
to say, if you want to creat a more efficient rotovap process, bigger is generally better. As bigger flask have
bigger surface, which will make more of the flask be contact with water / oil bath, makes your sample more
uniform heating and faster.
In brief, larger flask will make the surface area of the sample exposed to air inside the falsk, finally makes the
very adaptable when processing cbd oil. Hotsale 100L rotary evaporator in hemp oil industry.
2.Increasing the evaporation flask rotating speed.
Increasing the flask rotating speed can improve the heat transfer to the flask and solvent, and increase
the surface area of the liquid inside the flask, so as to improve the evaporation rate.
Most customers always think the faster of rotating speed is better, but pls pay attention to the turbulence
of liquid, too faster speed not only can actually lower the efficient of evaporation,but also will cause
premature wear on your rotary evaporator.
The distillation flask is recommend to rotate from 150-200rpm (will be different for different capacity rotary
evaporators, and different sample), which will increases the solution surface area and enhances the solvent
3.Increasing the rotary evaporator's bath temperature.
CBD extracts itself require lower water bath temperature to minimize thermal decomposition, so it's
important not to increase the rotary evaporator bath temperature too higher & fast.
And the evaporation temperature is also depending on the chiller you're using, if you set too high temp.
will increase your evaporation rate to a point that the chiller can't keep up.
So we could heating the rotary evaporator bath slowly, 30-40℃ is recommend, the condenser temperature
which controlled by a low temperature circulating bath chiller, could set to -10℃ to 0℃.
4.Increasing the vacuum level.
Using vacuum pump could reduced the pressure in the rotary evaporator, lower the boiling point of the solvent /
ethanol, you could finish solvent removing / ethanol recovering process at a lower temperature, protect the cbd
oil material, and could fast your evaporation rate.
Contact email@example.com to discuss more, or get technical support free and in time.