How to do rotary evaporator troubleshooting?
Rotary evaporator is widely used to remove solvent in cbd oil distillation process, and economical
and idea instrument for concentration, purification, evaporation, etc.
How to do simple troubleshooting by self when you meet the common problems ofrotary
1. The motor of rotary evaporator does not turn
- Connect the broken line and replug.
- Update the fuse or confirm that there is no abnormality in the power supply.
- Update the circuit board or electrical control box.
- Connect the ground wire or change the work location.
- Exclude according to the inverter manual P33.
2. Rotavapor's water bath / oil bath is not heated
- Detect the power supply is correct.
- Change the heating coil. Replace the solid state relay or relay board.
- Change the temperature controller. Detect the probe wiring or update the probe.
3. Bath pot temperature detection
- Change the temperature controller.
- Replace the solid state relay or relay board.
- Change the probe or temperature controller of vacuum rotary evaporator.
4. Vacuum can't pump
- Empty the solvent and try the empty flask of rotary evaporators.
- Vacuum oil pump for oil (water) changing, cleaning and maintenance.
- Detect and eliminate piece by piece along the vacuum line.
- Reassemble, clean the glass wool sandwich panel, apply vacuum grease, and tighten the flanges.
If your problems of rotary evaporator have not included in above situations, or you need some spare
parts of rota evaporator, please feel free contact email@example.com to get technical
support from rotary evaporator manufactures directly.